Axess RFS Nasal Mask System Instructions for Use
Instructions for Use Axess RFS Nasal Mask System
Scavenging Setup And Operation: (For more details, see scavenger connection instructions, 28626.) RFS™: Connect vacuum connector direct, no adapter needed. Adjust to top of green zone. Crosstex®/Accutron™ Accu-Vac™: Contact Customer Service for Adapter Kit. Open to max flow, disregard LED color. Matrx™, Remote: Use adapter. Adjust flow needle to top of acceptance zone. Porter® AVS: Use adapter. Set vacuum flow to green zone prior to connecting circuit. Ball will hit top of tube when connected to circuit. Warning! Only one vacuum controller may be connected at one time.
Crosstex®/Accutron ™ RFS ™
Crosstex®/Accutron ™ Accu-Vac ™
Matrx ™ Remote
Porter ® AVS
Patient Prep Procedure: - Select proper size mask (S, M, L).
- Attach mask to circuit per picture below. - Fit over nose and loop tubes over ears. - Tighten bolo slide under chin.
Nitrous Oxide Procedure: - Set total flow to 5.0 lpm Oxygen (Nitrous Oxide at 0 lpm or 0%). - Gradually increase Nitrous Oxide flow (%) until desired clinical effect is achieved.
- If desired clinical effect is not reached, increase Total Flow as needed up to maximum flow. - At end of procedure, turn off nitrous oxide and continue 100% oxygen for minimum of 5 minutes.
Accutron, Inc. 1733 W Parkside Lane Phoenix, AZ 85027 USA +1 800.531.2221 www.accutron-inc.com
MT Promedt Consulting GmbH Altenhofstrasse 80 D-66386 St. Ingbert/Germany
Te. +49 (0) 6894-581020 Fax +49 (0) 6894-581021
Nitrous Oxide/Oxygen Sedation
Axess ® is a registered trademark of Crosstex International, Inc., a Cantel Medical Company. Accutron ™ is a trademark of Accutron, Inc., a subsidiary of Crosstex International, Inc. Porter ™ is a trademark of Parker-Hannifin.
Part No. 28555 - Rev. 01 - 11/2017
Page 2 of 2
Made with FlippingBook - Online catalogs